BioMed Electrodes is a Manufacturer of Ag/AgCl Electrodes
BioMed Products is a manufacturer of reusable “Sintered Silver-Silver Chloride Electrodes” (Ag/AgCl matrix) supplying EEG Cap manufacturers throughout the World with superior quality Sensors. We work closely with our clients to design custom electrodes that fit their needs and requirements.
Silver-Silver Chloride (Ag/AgCl) Electrodes are known for their low offset voltage, low noise and stability. Resusable and non-toxic, the electrodes can be cold sterilized and used indefinitely with minimal care.
BioMed Products, Inc is known for their EEG Electrode/Sensor superiority, our electrodes are used in EEG, ECG, EMG, ENG, EOG GSR and Evoked Potentials applications.
“Sintered” Silver Silver Chloride Electrodes- Reusable Biopotential EEG Skin Electrodes
The BioMed Electrode is manufactured through a proprietary process using the highest purity chemicals that results in a fine grain homogeneous mixture, no fillers or binders are used- only pure silver-silver chloride. The electrodes are then compressed into a 1mm thick sensor and “sintered” (heat treated) for strength and stability. Standard lead wires are PVC insulated multi-strand copper with internal Tinsel for additional strength. The result is a versatile and durable, waterproof assembly that is reusable and resurfacable. They are easily reshaped by drilling, cutting or sanding.
The BioMed electrode is 1mm thick and encapsulated in durable epoxy housings. Our electrodes come with 1m or 1.5m PVC insulated lead wires, terminated with a 1.5 DIN safety socket. Performance is significantly superior to gold and silver cup electrodes.
Following are typical values of key parameters between pairs of electrodes of the same size measured after prolonged soaking in 0.9% saline. These are median values from a large sample:
DC offset Voltage (biaspotential): 180 µV, Drift: 25 µV / h (at constant temperature) Noise: 1 µV P – P (0.1 – 1 (0.l – l kHz with 60 Hz filtered out) Polarization (counter EMF): 2 – 4µV at 0.1 µA
Never need chlording
Unlike plated or chlorided silver, the BioMed Ag/AgCl electrodes are homogeneous and solid throughout. Should the surface become damaged or contaminated, a new surface can be exposed to restore the electrode’s original performance.
Reusable Sintered Silver Silver Chloride 150 Snap Electrode- 4mm$20.00 Add to cart
Reusable Sintered Silver Silver Chloride 200 Snap Electrode-8mm$25.00 Add to cart
Reusable Sintered Silver Silver Chloride Electrode 12MM Sensor 20cm Leadwire BME-12W$21.00 Add to cart
Reusable Sintered Silver Silver Chloride Fixed Wire Electrode 8mm sensor 20cm leadwire BME 8W$19.00 Add to cart
Reusable Sintered Silver Silver Chloride Fixed Wire Electrodes 4MM Sensor 20cm Leadwire BME 4W$17.00 Add to cart
Reusable Sintered Silver Silver Chloride Fixed Wire Electrodes 4MM Sensor – 1 Meter Leadwire BME-150$20.00 Add to cart
Reusable Sintered Silver Silver Chloride 14MM Sensor 1 Meter Leadwire Electrode BME-14$24.00 Select options
Reusable Sintered Silver Silver Chloride Electrode 8mm Sensor 1 Meter Leadwire BME-8$18.00 Select options
Reusable Sintered Silver Silver Chloride Electrode 6mm Sensor 1 Meter Leadwire BME-6$17.00 Select options
Reusable Sintered Silver Silver Chloride Electrode 12mm Sensor 1 Meter Leadwire BME-300LP$26.00 Select options
Reusable Sintered Silver Silver Chloride Electrode 8mm Sensor 1 Meter Leadwire BME-200LPX$23.00 Select options
Reusable Sintered Silver Silver Chloride Electrode 8mm Sensor 1 Meter Leadwire BME-200LP$22.00 Select options
Major EEG Cap manufacturers throughout the world rely on the BioMed Electrode to obtain the quality results they require.
Handling and Mounting Electrodes
Pellets & Discs
Avoid electrode contamination do not touch with bare fingers, do not mount in direct contact with metals. Do not store with metals such as iron or aluminum. Store in a dry, dark container.
Shaping of electrode should be done after it has been mounted in epoxy or supporting medium. When working with bare disks and pellets hold wire with tweezers at the base, avoid repeated bending of silver wire.
Epoxy adhesives are recommended for mounting electrodes on apparatus. Encapsulants should have the following properties. Plastic rings or tubes made of epoxy, acrylics or glass are recommended to support electrodes
Paste like, good adhesion, resistance to chloride solutions. The epoxy adhesive used should withstand water emersion and have good adhesion to glass or plastic.
Darkening of Electrode
Silver Chloride is light sensitive. Exposure to light will cause darkening to the outer surface, this will not affect the performance of the electrode. The
darkened surface can be removed if desired by light sanding with a fine non-metallic sandpaper. Cleaning with chemical agents should not be attempted.
To achieve stable potentials, immerse the electrodes (in pairs with their leads shorted to each other) in a selected electrolyte for 2-4 hours. Avoid touching or contaminating the bare electrodes surface, otherwise offset will increase.
Silver-silver chloride electrodes will withstand temperatures us to 200◦ C. Lower temperatures may be needed as adhesive or encapsulating materials will be damaged by high temperatures.
After most electrode applications rinsing with distilled water or isopropyl alcohol will suffice. Dirty or corroded electrodes can be re-surfaced by sanding off the electrode with a fine sand paper or pumice cleaner. Do not use emery cloth. Mounted electrodes can be cleaned by scrubbing with a brush and detergent water. Follow by rinsing thoroughly with water, drying and storing as recommended. Acetone or other strong solvents should not be used.
If sterilization is required, use ETO gas. Chemical disinfectants containing mercury, phenols, bromide, iodine, zinc, tin or organo-metallic compounds should not be used. Epoxy and cable insulation will not withstand steam sterilization.